ASTM D is a significantly updated version of the Standard Test Method for Determination of Thermal Conductivity of. Soils and Rock by Thermal Needle . ASTM D – Standard Test Method for Determination of Thermal Conductivity of Soil and Soft Rock by Thermal Needle Probe Procedure. This standard is. Buy ASTM D TEST METHOD FOR DETERMINATION OF THERMAL CONDUCTIVITY OF SOIL AND SOFT ROCK BY THERMAL.
|Published (Last):||17 September 2017|
|PDF File Size:||8.21 Mb|
|ePub File Size:||7.33 Mb|
|Price:||Free* [*Free Regsitration Required]|
Reliable testing depends on many factors; Practice D provides a means of evaluating some of those factors. In TP08 the reference junction of the ashm is located in the base. Mini VLQ now available. It is recognized also that it would be unwise, because of the standardization of this test method, to resist in any way the further development of improved or new methods or procedures by research workers.
In general a NSSP consists of a heating wire, representing a perfect line source, and a temperature sensor capable of measuring the temperature at this source. The sample size is not critical, as long as a 20 mm radius around the needle is covered. Users of this test method are cautioned that compliance with Practice D does not in itself assure reliable testing. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the sstm of regulatory limitations prior to use.
Copyright Compu-tecture, Inc.
Special models have been developed for in-situ field experiments. Because of the relative importance of the thermal equilibrium between base and qstm, an optional container can be ordered with TP Hukseflux is specialised in NSSP design. The new Mini VLQ is a 50 mm long needle probe with a diameter of 2 mm for small samples. Chosing an evaluation method is a compromise: Mathematically more complex approximations describe the form of the temperature curve more closely than a straight line.
Downloads Download Product Manual. The thermal conductivity of the sample is calculated from the temperature rise with time registered by the probe. Remember me for one month. When using s5334-08 modified method, part of the heat generated during measuring penetrates into the probe body instead of the sample material.
Min 20 mm radius. This test asrm is suitable only for isotropic materials. Active view current version of standard. See also Solutions for measurement and control: Link to Active This link will always route to the current Active version of the standard. This test method is applicable for both undisturbed and remolded soil specimens and soft rock specimens. This includes compressible materials like soil samples which cannot be tested with the already available Mini HLQ for plane surfaces.
See the TP02 manual or inquire at Hukseflux. Turn key systems are offered for sale: Usually this is achieved by clamping the base into the same material atsm that holds the sample. However, care must be taken to prevent significant error from: Permanent installation in soils: TK04 can use both methods by simply changing the connected probe. For some applications however the requirements regarding the sample size are such that the needle length and diameter of the TP02 are too large.
This test method is applicable for both undisturbed and remolded soil specimens and soft rock specimens. TP08 is a probe that offers the possibility to perform a practical and fast measurement of the thermal conductivity or thermal resistivity of the medium in which it is inserted at a high accuracy level.
Min volume 80 ml. Work Item s – proposed revisions of this standard.
TP08 – Hukseflux USA
Click here to download full list of books. These are done without need of reference materials. This model offers optimal measurement accuracy by a combination of its design features. A separate brochure is available. Optimal accuracy Temperature sensor in the base.
It can be used where standard size probes are too large for the available sample dimensions. More details on the science: TK04 comes with a reference material for checking astn operating condition of instrument and probes from time to time. As hard or brittle sample materials are often difficult to prepare for inserting a conventional needle probe, our modified method uses a needle embedded in the underside of a disk-shaped probe body which is just placed on top of astj sample surface probe for plane surfaces.
They yield more accurate results, and they can be used to detect disturbances in the data by comparing the measured temperature curve to the theoretical curve form. The slower the source temperature rises, the higher is the thermal conductivity of the sample material.
TK04 uses the well-established transient line source method widely known as needle probe method according to ASTM D The most common approximation method calculates thermal conductivity fron the slope of a straight line fitted to the measuring data.
Needle Probe Method According to ASTM D5334-08
This method may also be used for specimens containing moisture. If the results are to be reported as d5334-80 been obtained by this test method, then all pertinent requirements prescribed in this test method shall be met.
The Mini VLQ has a measuring range of 0. Enter your personal account email address to request a password reset: If you like to setup a quick demo, let us know at support madcad.